Please copy and paste this embed script to where you want to embed

Orifice Plate Calculator Pressure drop Calculations

1 de 4

Ads

Orifice plate calculator pressure drop calculation

JMB Zeppelin -Transp Pneu www.jmbz.com.br Equipment Transp Dosar Armazenar Po Plastico,Cimento,PVC,Cinza,C Black,

Fuel FlowMeters Low cost transit-time flowmeter for liquid. Noninvasive, no maintenance

Pressure Measurement Tekscan.com/Pressure-Measurement Measure Pressure Distribution with our Thin, Flexible Sensors/Systems.

Shenitech.com/FuelFlowMeters

Pressure measurement Long-term stable & robust sensors process & differential pressure

Multiphase Flow Meter www.jiskoot.com Multiphase meter for well test and reservoir management Flow Switch & Sensors www.thomasprod.com Flow & Level Sensor, Switch & More! Durable, Reliable & Custom Products

www.vega.com

Water Hammer Analysis Fast, Accurate Software - Large Network Capabilities Free Demo

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

Orifice plate pressure drop calculator is used for flow rate measuring in pipe systems. With orifice plate, pressure drop is created. Based on the magnitude of pressure drop, flow rate can be calculated. The pressure drop calculation formula for the pipe is explained in more detail below .Practical for larger tube diameters and for dirty fluid where other methods are impractical.

kypipe.com/surge

Flow Rate Measurement The Pitotless Nozzle™ FM Apporved Accuracy

Measure pressure drop from position 1 to position 2 and calculate flow rate and more with this easy to use flow calculator .Pressure drop formula calculation is published. It can be used as liquid, gas or air flow meter and the results can be checked against the textbook formula to gain confidence in using this easy to use orifice plate pressure drop /flow rate calculation . Flow rate can be expressed in metric or imperial units.

www.hosemonster.com/pitotless.html

Ultra Low Liquid Flow... Sensors and micro flow meters unique performance, small & light

Types of Flowmeters » Air Flow Meter » Ultrasonic » Smart Meters » Thermal Mass » Mass Flow Controller (MFC) » Coriolis Mass » Rotameter » Positive Displacement PD Meter » Vortex shedding » Multiparameter Mass Vortex » Multiphase » Turbine » Gas Flowmeter » Compressed Air Meter » Calorimetric » Compound Meter » Electromagnetic » Flumes & Weirs » Insertion Flow Meter » Open Channel » Orifice plate » Paddle Meter » Variable Area » Venturi » Doppler » Transit-Time » Portable Flow meter » Low Flow Measurement » Laminar Mass » Medical » OEM » Differential Pressure » Flow Meter

www.sensirion.com

Flow Indication & Control » Straighteners / Conditioners » Static Mixers » Vaporizer Systems » Sight Glass Indicators » Flow Switch » Indicator Panels » Watercut meters » Sensors » Nozzles » Crude Oil Mixing » Leak Detection » Air Eliminators » Automatic Meter Reading » Net Oil Computer » Water Meters » Wind Meters » Test and Calibration Centers » Flow Bench » Hydraulic Flow Switch » Flowmeters

(requires Java Runtime Environment http://java.com/en/download/index.jsp )

Explanation of used values

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

2 de 4

D1 D2 p1 p2 p1 - p2 ni mi T rho R kappa

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

diameter of tube throat diameter upstream pressure downstream pressure pressure drop through orifice kinematic viscosity dynamic viscosity upstream temperature upstream density gas constant isentropic coefficient

Resulting values Q G V1 V2 ReD e C

volumetric flow rate mass flow rate upstream velocity throat velocity upstream Reynolds number expansion coefficient coefficient of discharge

Infrared Boiler Thermomet Infra-View Thermometers are remote detectors to penetrate any boiler www.Infra-View.com

Mass Flow Instruments Thermal and Coriolis mass flow meters & controllers for gas/liquid www.bronkhorst.com

Siemens answers: Efficient energy supply The world's largest gas turbine. www.siemens.com/answers

Theory Calculation of flow rate using orifice plate calculator is for incompressible flow, based on the Bernoulli principle:

where is: p - pressure rho - density V - velocity g - gravitational constant (9.81 m/s2) z - geodetic height Assumption that pressure lost is negligible (pressure drop is obvious and included with coefficient of discharge which is introduced bellow):

Galperti Manufacturer A Variety of Flanges from 2" to 96" Highest Quality Flange Products www.Galperti-am.com

ShaftDesigner The Shaft Calculation Software for Modelling, Alignment and Vibration www.shaftdesigner.com

and:

and if velocities substituted with flow rate:

where is: Q - volumetric flow rate D - diametar Pressure drop through the orifice because of velocity increase can be calculated as follows:

or:

Expressing flow rate from the previous equation leads to:

Substituting:

flow rate can be determined as:

where is: C - coefficient of discharge e - expansion coefficient

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

3 de 4

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

Coefficient of discharge can be calculated using following equation (ISO):

where is: beta - diameter relation D2/D1 ReD - Reynolds number which can be calculated as follows:

where is: ni - kinematic viscosity mi - dynamic viscosity

L1 and L2 are functions on tap type and it is: L1=L2=0 for corner taps L1=1 L2=0.47 for D & D/2 taps L1=L2=0.0254/D D[m] for 1" taps

Expansion coefficient e can be calculated (for gases only):

where is: kappa - isentropic coefficient; kappa = 1.4 for air and other two atom gas molecules Other values are calculated using following equations: mass flow:

velocities:

If flowing fluid is gas, then it is considered as incompressible and ideal. Equation for ideal gas:

can be used for calculation of temperature T:

as well as density rho:

where R is gas constant (R=287 J/kgK for air).

Appliance Orifice plate calculator can be used for both liquids and gases. Fluid is considered as incompressible, so density (rho) and temperature (T) are constant through tube. Also, gas is considered as ideal. Units of measure are both in SI and English system. Orifice plate calculator can be used for calculation of: volumetric flow - Q mass flow - G velocity on inlet, bigger diameter - V1 velocity on smaller diameter - V2 Reynolds number on bigger diameter - ReD For calculation of those values, necessary values for input are: inlet, bigger diameter - D1

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

4 de 4

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

smaller diameter - D2 inlet pressure - p1 either pressure on smaller diameter - p2, or pressure drop p1-p2 either kinematic viscosity - ni, or dynamic viscosity - mi for gases only: either temperature - T, or density - rho gas constant - R for liquids: density - rho Beside four values (Q, G, V1, V2), which calculation is main purpose of calculator, values that are not defined by user are determined in process of calculation (for example: if pressure p1 and temperature T are specified for gas flow value for rho is calculated together with four main values (Q, G, V1, V2)). Also necessery coeffients: expansion - e and discharge - C are calculated and results are shown. Orifice plates and metering runs are popular because of low cost . Orifice plates combined with differential pressure transmitters are now commonly supplied as complete flow measurement packages. Multiport pressure transmitter manifold valves gives options to operators for e.g to isolate , equalize and vent the differential pressure transmitter. Orifice plates and differential pressure measurement devices are used to measure the flow of fluids such as steam, air, gas and water. In natural gas transmission a familiar type of orifice plate is the senior orifice plate. This special design allows operators to remove and inspect the orifice plate during normal pipeline operation. A metering run, flow straightener may have to be installed to correct the flow profile in custody transfer applications or orifice plate applications that require higher accuracy.

22/01/2011 10:07

View more...
1 de 4

Ads

Orifice plate calculator pressure drop calculation

JMB Zeppelin -Transp Pneu www.jmbz.com.br Equipment Transp Dosar Armazenar Po Plastico,Cimento,PVC,Cinza,C Black,

Fuel FlowMeters Low cost transit-time flowmeter for liquid. Noninvasive, no maintenance

Pressure Measurement Tekscan.com/Pressure-Measurement Measure Pressure Distribution with our Thin, Flexible Sensors/Systems.

Shenitech.com/FuelFlowMeters

Pressure measurement Long-term stable & robust sensors process & differential pressure

Multiphase Flow Meter www.jiskoot.com Multiphase meter for well test and reservoir management Flow Switch & Sensors www.thomasprod.com Flow & Level Sensor, Switch & More! Durable, Reliable & Custom Products

www.vega.com

Water Hammer Analysis Fast, Accurate Software - Large Network Capabilities Free Demo

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

Orifice plate pressure drop calculator is used for flow rate measuring in pipe systems. With orifice plate, pressure drop is created. Based on the magnitude of pressure drop, flow rate can be calculated. The pressure drop calculation formula for the pipe is explained in more detail below .Practical for larger tube diameters and for dirty fluid where other methods are impractical.

kypipe.com/surge

Flow Rate Measurement The Pitotless Nozzle™ FM Apporved Accuracy

Measure pressure drop from position 1 to position 2 and calculate flow rate and more with this easy to use flow calculator .Pressure drop formula calculation is published. It can be used as liquid, gas or air flow meter and the results can be checked against the textbook formula to gain confidence in using this easy to use orifice plate pressure drop /flow rate calculation . Flow rate can be expressed in metric or imperial units.

www.hosemonster.com/pitotless.html

Ultra Low Liquid Flow... Sensors and micro flow meters unique performance, small & light

Types of Flowmeters » Air Flow Meter » Ultrasonic » Smart Meters » Thermal Mass » Mass Flow Controller (MFC) » Coriolis Mass » Rotameter » Positive Displacement PD Meter » Vortex shedding » Multiparameter Mass Vortex » Multiphase » Turbine » Gas Flowmeter » Compressed Air Meter » Calorimetric » Compound Meter » Electromagnetic » Flumes & Weirs » Insertion Flow Meter » Open Channel » Orifice plate » Paddle Meter » Variable Area » Venturi » Doppler » Transit-Time » Portable Flow meter » Low Flow Measurement » Laminar Mass » Medical » OEM » Differential Pressure » Flow Meter

www.sensirion.com

Flow Indication & Control » Straighteners / Conditioners » Static Mixers » Vaporizer Systems » Sight Glass Indicators » Flow Switch » Indicator Panels » Watercut meters » Sensors » Nozzles » Crude Oil Mixing » Leak Detection » Air Eliminators » Automatic Meter Reading » Net Oil Computer » Water Meters » Wind Meters » Test and Calibration Centers » Flow Bench » Hydraulic Flow Switch » Flowmeters

(requires Java Runtime Environment http://java.com/en/download/index.jsp )

Explanation of used values

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

2 de 4

D1 D2 p1 p2 p1 - p2 ni mi T rho R kappa

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

diameter of tube throat diameter upstream pressure downstream pressure pressure drop through orifice kinematic viscosity dynamic viscosity upstream temperature upstream density gas constant isentropic coefficient

Resulting values Q G V1 V2 ReD e C

volumetric flow rate mass flow rate upstream velocity throat velocity upstream Reynolds number expansion coefficient coefficient of discharge

Infrared Boiler Thermomet Infra-View Thermometers are remote detectors to penetrate any boiler www.Infra-View.com

Mass Flow Instruments Thermal and Coriolis mass flow meters & controllers for gas/liquid www.bronkhorst.com

Siemens answers: Efficient energy supply The world's largest gas turbine. www.siemens.com/answers

Theory Calculation of flow rate using orifice plate calculator is for incompressible flow, based on the Bernoulli principle:

where is: p - pressure rho - density V - velocity g - gravitational constant (9.81 m/s2) z - geodetic height Assumption that pressure lost is negligible (pressure drop is obvious and included with coefficient of discharge which is introduced bellow):

Galperti Manufacturer A Variety of Flanges from 2" to 96" Highest Quality Flange Products www.Galperti-am.com

ShaftDesigner The Shaft Calculation Software for Modelling, Alignment and Vibration www.shaftdesigner.com

and:

and if velocities substituted with flow rate:

where is: Q - volumetric flow rate D - diametar Pressure drop through the orifice because of velocity increase can be calculated as follows:

or:

Expressing flow rate from the previous equation leads to:

Substituting:

flow rate can be determined as:

where is: C - coefficient of discharge e - expansion coefficient

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

3 de 4

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

Coefficient of discharge can be calculated using following equation (ISO):

where is: beta - diameter relation D2/D1 ReD - Reynolds number which can be calculated as follows:

where is: ni - kinematic viscosity mi - dynamic viscosity

L1 and L2 are functions on tap type and it is: L1=L2=0 for corner taps L1=1 L2=0.47 for D & D/2 taps L1=L2=0.0254/D D[m] for 1" taps

Expansion coefficient e can be calculated (for gases only):

where is: kappa - isentropic coefficient; kappa = 1.4 for air and other two atom gas molecules Other values are calculated using following equations: mass flow:

velocities:

If flowing fluid is gas, then it is considered as incompressible and ideal. Equation for ideal gas:

can be used for calculation of temperature T:

as well as density rho:

where R is gas constant (R=287 J/kgK for air).

Appliance Orifice plate calculator can be used for both liquids and gases. Fluid is considered as incompressible, so density (rho) and temperature (T) are constant through tube. Also, gas is considered as ideal. Units of measure are both in SI and English system. Orifice plate calculator can be used for calculation of: volumetric flow - Q mass flow - G velocity on inlet, bigger diameter - V1 velocity on smaller diameter - V2 Reynolds number on bigger diameter - ReD For calculation of those values, necessary values for input are: inlet, bigger diameter - D1

22/01/2011 10:07

Orifice Plate Calculator Pressure drop Calculations

4 de 4

http://www.flowmeterdirectory.com/flowmeter_orifice_calc.html

smaller diameter - D2 inlet pressure - p1 either pressure on smaller diameter - p2, or pressure drop p1-p2 either kinematic viscosity - ni, or dynamic viscosity - mi for gases only: either temperature - T, or density - rho gas constant - R for liquids: density - rho Beside four values (Q, G, V1, V2), which calculation is main purpose of calculator, values that are not defined by user are determined in process of calculation (for example: if pressure p1 and temperature T are specified for gas flow value for rho is calculated together with four main values (Q, G, V1, V2)). Also necessery coeffients: expansion - e and discharge - C are calculated and results are shown. Orifice plates and metering runs are popular because of low cost . Orifice plates combined with differential pressure transmitters are now commonly supplied as complete flow measurement packages. Multiport pressure transmitter manifold valves gives options to operators for e.g to isolate , equalize and vent the differential pressure transmitter. Orifice plates and differential pressure measurement devices are used to measure the flow of fluids such as steam, air, gas and water. In natural gas transmission a familiar type of orifice plate is the senior orifice plate. This special design allows operators to remove and inspect the orifice plate during normal pipeline operation. A metering run, flow straightener may have to be installed to correct the flow profile in custody transfer applications or orifice plate applications that require higher accuracy.

22/01/2011 10:07

Thank you for interesting in our services. We are a non-profit group that run this website to share documents. We need your help to maintenance this website.

To keep our site running, we need your help to cover our server cost (about $400/m), a small donation will help us a lot.